Skip to 0 minutes and 12 secondsWe're in the Mechanics of Solids Laboratory, with its machines for testing materials for strength and stiffness. They can test components, too, like a rubber bearing for a bridge support. A bridge bearing would need a large machine - we can use this smaller one. We're going to test this wire strap. The machine will give us a plot of load versus deflection, right up until it breaks. Each cable you'll specify for suspending the loudspeaker will be like this. We'll start the test.
Skip to 1 minute and 2 secondsThe strap failed at the clamp. We expected that. The cable for suspending the loudspeaker would probably fail at the clamp or swage, too. From the chart, the maximum load before it broke - it's ultimate strength was 3.2 kilonewtons. The ultimate load specified for the wire was 3.7 kilonewtons. It probably failed before it reached this value in the test because of the clamp. See how the clearance in the system is gradually taken up. Then the curve becomes approximately linear before decreasing in the slope, until finally the wire fails suddenly. There are seven strands in the wire, but only four strands broke. After a while the others took up the load again, but it had definitely failed.
Skip to 2 minutes and 0 secondsDifferent materials produce different shapes of curves. For example, here's our chart from week one of load versus deflection for a rubber band. Now to this week's experiments, force transducers. In this experiment, you will learn how an elastic component can be used as a force transducer. A transducer is a device that converts something we can't measure into something else that we can. This spring balance is a transducer. When you load it up the spring stretches, and we measure the stretch using a calibrated scale that reads directly in force units. The load applied by the test machine is measured by a transducer.
Skip to 2 minutes and 49 secondsIt uses a spring element, too, but in this case the deflection is tiny and is measured electronically by an electrical strain gauge. But the principle is the same. It has to be calibrated, too. This luggage scale also uses a strain gauge. Some bathroom and kitchen scales do, too. They all have to be calibrated in some way. In this experiment, we'll make a matched pair of transducers out of our chains of rubber bands. We'll use our two-bands chain, so we can measure moderate loads more accurately. Then, in the next experiment, we'll use them to investigate how forces add.
Skip to 3 minutes and 30 secondsOf course, we'll have to calibrate our transducers, and for this we'll need to find the weight of the pans in terms of washers - because one washer is our unit of weight. Let's get started and weigh the pans. We'll load the chain of bands until the initial stretch is taken up, then hang the second pan from the chain and note the extension. Then we'll take the pan off and add washes until the extension is the same as with the pan. We'll repeat this for the other pan. To get a calibration curve we'll load up a transducer in stages, recording weight and extension. We've done this already for one chain of bands.
Skip to 4 minutes and 14 secondsAll we need to do for that one is add the weight pan into the total load, and plot total weight against length. Then, we will repeat this for a second chain of rubber bands. Here are the calibration curves. They're not straight lines, so we'll have to read off the force for each extension measurement. Commercial force transducers are more linear, but even they might use a calibration curve for precise work. We've got our transducers. In the next video, we'll use them to see how forces add.
Experiment: Force transducers
Forces generated in an earthquake can destroy entire cities, so it might seem odd to say that they are just an abstract construct. More on that in the analysis activities.
For now, we need a way of detecting a force, measuring how big it is and specifying the direction it acts in. For this we need a transducer - a device that converts something we can’t measure into something else that we can.
Our force transducers will be based on the chain of rubber bands that we made during the experiment in Week 1.
If you attempt the experiment, take a photo and upload it to our Through Engineers’ Eyes Padlet wall. You can include a link to your photo in the comments for this step (click on your post on the Padlet wall and then copy the web address).
If you don’t have access to a camera, we encourage you to still do the activity and write a description of your experiment on the Padlet wall.
For a guide to using Padlet, please see FutureLearn’s social media help page.
Once we have our transducers we’ll use them to find out how forces add.
If you didn’t do the Week 1 experiments it’s not too late to do them now. It would be a good investment because you can use the transducers this week and next.
You can download instructions to the experiment in the Downloads section below.